Our mission is to become a worldwide reference for education in the field for all professionals involved in the process to disseminate knowledge & skills of Acute Cardiovascular Care.
Our mission is to promote excellence in clinical diagnosis, research, technical development, and education in cardiovascular imaging.
Our mission is to promote excellence in research, practice, education and policy in cardiovascular health, primary and secondary prevention.
Our mission is to reduce the burden of cardiovascular disease through percutaneous cardiovascular interventions.
Improving the quality of life and reducing sudden cardiac death by limiting the impact of heart rhythm disturbances.
Our mission is to improve quality of life and longevity, through better prevention, diagnosis and treatment of heart failure, including the establishment of networks for its management, education and research.
The ESC Working Groups' goal is to stimulate and disseminate scientific knowledge in different fields of cardiology.
The ESC Councils' goal is to share knowledge among medical professionals practicing in specific cardiology domains.
BACKGROUND: The computed tomographic (CT) attenuation of coronary atherosclerotic plaque has been proposed as a marker for tissue characterization and may thus potentially contribute to the assessment of plaque instability.
OBJECTIVE: We analyzed the influence of reconstruction parameters on CT attenuation measured within noncalcified coronary atherosclerotic lesions.
METHODS: Seventy-two patients were studied by contrast-enhanced dual-source CT coronary angiography (330 millisecond rotation time, 2 x 64 x 0.6 mm collimation, 120 kV, 400 mAs, 80 mL contrast agent intravenously at 6 mL/s), and a total of 100 distinct noncalcified coronary atherosclerotic plaques were identified. Image data sets were reconstructed with a soft (B20f), medium soft (B26f), and sharp (B46f) reconstruction kernel. With the medium soft kernel, image data sets were reconstructed with a slice thickness/increment of 0.6/0.3 mm, 0.75/0.4 mm, and 1.0/0.5mm. Within each plaque, CT attenuation was measured. RESULTS: Mean CT attenuation using the medium soft kernel was 109 +/- 58 HU (range, -16 to 168 HU).
Using the soft kernel, mean density was 113 +/- 57 HU (range, -13 to 169 HU), and using a sharp kernel, mean density was 97 +/- 49 HU (range, -23 to 131 HU). Similarly, reconstructed slice thickness had a significant influence on the measured CT attenuation (mean values for medium soft kernel: 102 +/- 52 HU versus 109 +/- 58 HU versus 113 +/- 57 HU for 0.6-mm, 0.75-mm, and 1.0-mm slice thickness).
The differences between 0.75-mm and 0.6-mm slice thickness (P = 0.05) and between medium sharp and sharp kernels (P = 0.02) were statistically significant.
Our mission: To reduce the burden of cardiovascular disease
© 2017 European Society of Cardiology. All rights reserved