Our mission is to become a worldwide reference for education in the field for all professionals involved in the process to disseminate knowledge & skills of Acute Cardiovascular Care.
Our mission is to promote excellence in clinical diagnosis, research, technical development, and education in cardiovascular imaging in Europe.
Our mission is to promote excellence in research, practice, education and policy in cardiovascular health, primary and secondary prevention.
Our mission is to reduce the burden of cardiovascular disease in Europe through percutaneous cardiovascular interventions.
Our mission is to improve the quality of life of the population by reducing the impact of cardiac rhythm disturbances and reduce sudden cardiac death.
Our mission is to improve quality of life and longevity, through better prevention, diagnosis and treatment of heart failure, including the establishment of networks for its management, education and research.
The ESC Working Groups' goal is to stimulate and disseminate scientific knowledge in different fields of cardiology.
The ESC Councils' goal is to share knowledge among medical professionals practising in specific cardiology domains.
OUR MISSION: TO REDUCE THE BURDEN OF CARDIOVASCULAR DISEASE
Lecturer: Doctor Marco GotteThe aims of this first lecture are:- create awareness for a (novel) growing clinical problem: Clinicans are confronted with an increasing number of patients with a device, and an increasing clinical need for a MRI scan- explain the (technical reasons of) interaction between MRI and devices (and vice versa)- point out the effects/hazards of MRI scanning on pacemaker functioning- provide a strategy to overcome the MRI and device interaction and to minimize risk of scanning in (conventional) pacemaker patients- to pay attention to labelling of devices
Lecturer: Professor Richard Sutton
My talk will be about special aspects of MRI in patients with Devices and those who may receive them:
- Presence of redundant leads in the body is a very strong contraindication for MRI- How to persuade Clinical Radiologists that a scan is possible with reasonable precautions in patients who have received a MRI compatible device- The same process is also necessary for Clinical Neurologists as they are the most frequent requesters of MRI scans and many currently do not appreciate the technical progress recently made in Devices- Cost of Devices and impact on the budget of the implanting centre- If all patients cannot receive a MRI compatible Device who should receive?- Parts of the body that can be scanned with each approved Device and even if the isocentre of the scan is remote from the scan target what can be learnt by the CMR specialist or Radiologist- What Devices without MRI compatibility can be scanned? What precautions are necessary? ECG and Oximetry monitoring in all.
Subject: Old leadsNew MRI safe devices and old leads. How should this be handled?R.S: Old leads are a problem. A MRI conditional device will help but the indications for MRI must be strong and the precautions taken also very strong. Abandoned leads are an absolute contraindication for MRI.
Subject: Monitoring of the patient during the mri scanYou have told that the mri scan can disturb the ECG monitoring. During an MRI scan, don't we need to monitorate the electrical activity of the patient to synchronise images? R.S: ECG Monitoring is possible. The equipment is not cheap and the tracings available are imperfect but for CMR an ECG is obligatory. The imperfections are such that pulse oximetry is required in addition.
Subject: MRI in pt with PPMwould you recommend a pt with a non MRI PPM to have a MRI scan done? RS: Yes but the indications for MRI must be strong and preferably the patient should not be pacemaker dependent and the device and lead manufactured after 2000. If these criteria are not met the risk is higher for patient and pacing system.
Subject: MRI in PPM ptwhat is the risk of having a MRI before 6 weeks after a ppm implant? What the risk of a patient with Biotronik PPM to have a full body MRI?R.S:1. Six weeks are chosen to allow the leads to gain full stability in the heart by endotheliailisation. There is at least a theoretical danger of lead displacement by torque in those first 6 weeks.2. The risks of a full body scan in a device that is not approved for one are several. A: There will be an arrhythmia or asystole. B. There will be a great threshold rise on lead or leads requiring reprogramming or reoperation. C. There will be damage to the generator and D. If there are problems and you are sued, you are not covered at all against the legal risks
Subject: PM patientsIn patient with PM, swiched of the PM,,, MRI is totaly safe?R.S: No, the system is still subject to all the problems of MRI except that the device will not trigger an arrhythmia. Subject: non-MRI CEIDSCan we still do MRI in patients who do not have MRI safe CEIDSR.S: Yes, but a strong indication for MRI is required, plus preferably the device and leads should have been manufactured after 2000 and the patient not be pacemaker dependent. Strong precautions as discussed in the session are required. Subject: Perspective on IEDsWhat in your opinion is more cost effective? Removing and replacing existing devices with types that are safe with MRIs or simply investing more in research on how to better and more safely conduct MR scan with the already implanted devices?R.S: MRI is MRI and no amount of research will ameliorate the risks of MRI to CIED patients. The solution lies in the design of the devices and leads plus very careful precautions. MRI will develop in the future to pose new problems e.g., 3 and 7 Tesla machines which will pose new design problems on CIED systems. This is where the money needs to be spent. Subject: Device failure frequencyIs the safety problem of MRI in patients with cardiovascular implantable electronic device just hypothetical? Is there any large study showing a high frequency of device failure?R.S: It is not a hypothetical problem. Deaths of CIED patients in MRI have been recorded but for an up to date review of this I suggest to read Beinart R, Nazarian S. Magnetic resonance imaging in patients with implanted devices. J Cardiovasc Electrophysiol 2012; 23: 1040-1042. Subject: Insertable Cardiac MonitorsWhy Patients with a Reveal ICM implanted in sites other than the subcutaneous region of the chest are contraindicated for an MRI procedure?R.S: They are not contraindicated in any situation. ILRs, however, are implanted typically nearer the heart than other CIEDs. They make a large artefact and may adversely influence the interpretation of CMR as shown in my presentation.
Subject: old leadsNew MRI conditional devices and old leads. Safe with precautions?R.S: This combination requires the same precautions as if no part of the system is MRI conditional. May I suggest that it is better to avoid the term ‘safe’ as nothing in this field is safe. Whatever we do has risks and these have to be carefully weighed and the best decision made for each patient.
Subject: ImplantationPatients with renal failure often need an MRI because of contraindication to CT with iodinated contrast. Are these patients on the priorities list of the implantation of the new MRI-designed implantable cardiac devices?R.S: This is a valuable point and it is reasonable to give them priority
The safety of MRI scanning of pacemakers and ICDs: what are the critical elements of safe scanning? Ask me again at 10 000. Europace 2010; 12:, 915–917Wilkoff BL, Bello D, Taborsky M, Vymazal J, Kanal E, Heuer H, et al: EnRhythm MRI SureScan Pacing System Study Investigators. Magnetic resonance imaging in patients with a pacemaker system designed for the magnetic resonance environment. Heart Rhythm 2011; 8: 65-73.
Nazarian S, Hansford R, Roguin A, Goldsher D, Zviman MM, Lardo AC, Caffo BS, Frick KD, Kraut MA, Kamel IR, Calkins H, Berger RD, Bluemke DA, Halperin HR. A prospective evaluation of a protocol for magnetic resonance imaging of patients with implanted cardiac devices. Ann Intern Med. 2011;155: 415-424.
This educational programme is accredited for 1 CME Credit(s). Each participant should claim only those hours of credits that have actually been spent in the educational activity.
European Society of Cardiology
European Heart HouseLes Templiers2035 Route des CollesCS 80179 BIOT
06903 Sophia Antipolis, FR
The webinar is available thanks to an unrestricted educational grant from St Jude Medical.
© 2017 European Society of Cardiology. All rights reserved