Our mission is to become a worldwide reference for education in the field for all professionals involved in the process to dissemintate knowledge & skills of Acute Cardiovascular Care
Our mission is to promote excellence in clinical diagnosis, research, technical development, and education in cardiovascular imaging in Europe.
Our mission: To promote excellence in research, practice, education and policy in cardiovascular health, primary and secondary prevention.
Our goal is to reduce the burden in cardiovascular disease in Europe through percutaneous cardiovascular interventions.
Our Mission is "to improve the quality of life of the population by reducing the impact of cardiac rhythm disturbances and reduce sudden cardiac death"
To improve quality of life and logevity, through better prevention, diagnosis and treatment of heart failure, including the establishment of networks for its management, education and research.
Working Groups goals is to stimulate and disseminate scientific knowledge in different fields of cardiology.
ESC Councils goal is to share knowledge among medical professionals practising in specific cardiology domains.
OUR MISSION: TO REDUCE THE BURDEN OF CARDIOVASCULAR DISEASE
Prof. José-Luis Zamorano
Remote monitoring is available for pacemakers, internal cardio-defibrillator (ICDs), cardiac resynchronisation devices and other devices, such as the implantable Holter monitor. Of note, it can detect silent atrial arrhythmias or adverse events of inappropriate internal cardio-defibrillator therapies and all the while, reducing the number in-office visits. Review devices, recommendations and what to check for at visits (remote) in this article.
Interestling link: Cardiac Pacing and Resynchronisation therapy guidelines.
The rise in the number of elderly advanced heart failure patients has led to overcrowding of in-office follow-up management and increasing numbers of more performing but also more complex implantable cardiac devices that have added to the problem. Remote monitoring has helped to reduce the number of visits and allowed for more accurate follow-up and comes as a safe, cost-effective and comfortable alternative to face-to-face follow-up visits, especially useful for post-implant checks, where programming changes are rarely required (1,2,3).
On battery and lead parameters, physicians will access and check for significant electrophysiological events, such as arrhythmias, and review the therapy administered through implantable defibrillator devices (ICDs); in heart failure patients, they will look for transthoracic impedance and patient activity.As a complement to in-office cardiac device follow-up visits, remote monitoring that has developed and improve since 2001 is feasible, a safe, cost-effective (20) and increases patient and practitioner satisfaction. Remote monitoring for pacemakers, ICDs, cardiac resynchronisation devices and other devices, such as the implantable Holter monitor is available.
Home-monitoring (Biotronik) Germany; no landline necessary, travelling possible: This German system from 2001 uses a portable transmitter that sends information to a GSM network in Germany (Cardio Messenger). No landline telephone is necessary. Continuous day-long monitoring is provided, as is the possibility of transmission while traveling.Thirty-second electrograms are periodically sent for interpretation by a physician, and different alerts can be configured online. Significant events can be reported by fax, e-mail or text message in order to ensure optimal follow-up.Carelink (Medtronic) no landline necessary, only in patient's home country (Europe): This wireless system has been available in Europe since 2005. It requires an analogue landline phone and cannot transmit unless the patient is in his or her home country. One advantage is the automatic testing feature, including automatic atrial, right ventricular and left ventricular capture thresholds, which facilitates full remote follow-up.Remote visits should be scheduled on line by specialised health practitioners. Practitioners can also adjust different levels of alerts to individual patient characteristics. These alerts are transmitted to the physician by e-mail or text message. Another advantage is its coverage of transthoracic impedance monitoring in patients with dual chamber devices, potentially a useful option in heart failure management. Latitude (Boston Scientific): landline necessary but travelling possible. This system was first introduced in Europe in 2009. Its transmitter also uses an analogue landline that may be configured for use in more than one country. Its ability to transmit readings from wireless weight scales and blood pressure cuffs is a useful remote monitoring tool for the management of heart failure patients.
Event notifications can be configured individually, as in the Carelink system, based on different levels according to event severity. Another advantage is its ability to transmit information to different physicians, such as a general cardiologist and a general practitioner, with the resulting benefits in patient management. Merlin (St Jude Medical) communicates by radio in Europe. The wireless transmitter (Merlin@home) was introduced in Europe in 2008. The system communicates automatically by radio. It uses an analogue landline phone and can be fitted with a cellular adapter. Different events are transmitted to the physician by text message, fax or e-mail. It has the ability to leave reminders of in-office visits on the patient’s transmitter and to make automatic phone calls reporting that all parameters are normal (4 Europace 2009).
Most ICD monitoring systems feature these measures :
The lead system is an important component of all implantable devices. However, its complex electrodes are susceptible to defects, fractures, exposing patients to the risks of inappropriate therapy and failure of defibrillation when necessary or loss of pacing.Adverse clinical events related to lead dysfunction may be preceded by asymptomatic abnormal electrode characteristics, such as impaired lead impedance, oversensing of non-cardiac potentials or inappropriate increase in pacing rate.Remote monitoring can contribute to a prior diagnosis of lead dysfunction that can potentially avoid adverse events related to electrode fracture or dislodgement (5). Fig. 1: Recording from a Biotronik ICD (Home Monitoring system) device showing low amplitude of right ventricle R wave which could be related to lead dysfunction.
Clinical atrial fibrillation is a common arrhythmia responsible for 15% of ischemic strokes, however, in about 25% of such events no evident cause is detected. These are presumably related to non-clinical atrial fibrillation. When a lead is implanted in the atrium, asymptomatic episodes of atrial arrhythmias can be detected.Recently Healey et al. showed that in patients with cardiac devices and no clinical atrial fibrillation, subclinical atrial tachycardia was associated with an increased risk of ischemic stroke and concluded that subclinical atrial tachyarrhythmias often precede clinical atrial fibrillation. Routine use of remote monitoring systems could be useful for prior detection of silent episodes such as these, enabling the physician to start oral anticoagulation if indicated and to avoid thromboembolic events (6, 7, 8).Atrial arrhythmias may often go undetected (figure 2). Thus, a careful examination of electrograms is mandatory when revisions are performed remotely. Fig. 2. Panel A: Episode of silent atrial fibrillation detected by an ICD device. Panel B: Atrial tachycardia episode recorded, but not detected by a CRT-D device. Note that atrial activity (arrow) is regular, probably a typical atrial flutter, cycle length 200 msec. ICD: implantable cardiodefibrillator). CRT-D: cardiac resynchronisation therapy with defibrillator.
The main use of ICDs is to detect and treat potentially fatal ventricular arrhythmias in selected patients according to current guidelines (9, 10, 11).ICDs are promptly revised post implant and furthermore every three to six months, according to individual center availability (4). Nielsen et al. (12) published a prospective study of patients with a Biotronik ICD and Home Monitoring system with a mean follow-up of 10 months. At least one relevant event occurred in 41% of patients, mostly potentially lethal arrhythmias. Around 60% of these events occurred during the first month after the last in-office visit. Another study enrolling about 11,000 patients showed that the mean interval between the last face-to-face visit and a remote monitor-notified event was 26 days. That means a gain of 154 days compared with a six-month follow-up routine, and 64 days compared to a three-month follow-up routine (13).The results of these studies suggest the usefulness of a remote monitoring system for prior detection of clinically relevant events such as ventricular arrhythmias. Examples of different therapies are shown in figure 3. Fig.3: Panel A: Electrogram from a Medtronic ICD device obtained by Carelink system. Note the initial presence of ventricular tachycardia and non-effective anti-tachycardia pacing (blue arrow). Red arrow marks the persistence of a modified ventricular tachycardia after pacing. Panel B: Continuation of ventricular tachyarrhythmia (green arrow), terminated by an effective shock.
An ICD carrier has an increased risk for anxiety and depression disorders, mostly because of the shocks administered by the device. Therefore, avoiding inappropriate therapies in patients with cardiac devices may be a priority for the treating physician (14).A remote monitoring system could be useful for detecting lead dysfunction (such as dislodgement or fracture) or rapid atrial arrhythmias that could develop into inappropriate shocks. Examples of inappropriate therapies are shown in figure 4. Fig. 4: Panel A: Episode of rapid atrial fibrillation (red arrow) and inappropriate antitachycardia pacing (blue arrow). Panel B: Inappropriate shock (blue arrow) in a patient with rapid atrial fibrillation. Register from a Medtronic ICD device obtained by Carelink remote monitoring system.
Following up cardiac resynchronisation devices is challenging due the complexity of the device and the patient’s heart disease. Several potential causes of heart failure instability can develop into hospital admission or cardiovascular death. Added to the usual parameters reviewed above, certain remote monitoring systems offer additional measures, such as transthoracic impedance, physical activity, body weight or special alerts according to symptoms. Special attention should be payed by physicians to events such as:
Several studies have shown how prior detection of these events is possible with a remote monitoring system (15, 16).In 2012, Landolina et al. suggested that, compared with the usual practice, a proper remote monitoring strategy could reduce the number of emergency department visits (17).Indeed, the 2012 EHRA expert consensus on cardiac resynchronisation therapy includes remote follow-up visits in addition to clinic follow-ups, and it encourages patients to transmit if changes in symptoms appear (18).
Below are the recommendations offered by the European heart failure association in its expert document regarding pacemakers, ICDs and CRT(19):
Remote monitoring offers detection of adverse events even before symptoms appear. The physician is able to check most of the parameters recommended in the consensus document for proper device follow-up. The treating physician must search carefully for arrhythmic events and lead dysfunction.Prior detection of silent atrial arrhythmias offers the possibility of oral anticoagulation initiation in order to prevent ischemic strokes.Inappropriate ICD therapies are undesirable adverse events that have an important psychological impact on patients and can be detected by remote monitoring, for prompt management by the treating physician.Cardiac resynchronisation devices need special attention due to their complexity and due to the severity of the underlying cardiac disease. The expert consensus recommends remote monitoring as a valid complement to in-office visits.Further studies are needed in this field before remote monitoring is recommended as a complete alternative to physical follow-ups.
1 - Potential role of remote monitoring for scheduled and unscheduled evaluations of patients with an implantable defibrillator.Heidbüchel H, Lioen P, Foulon S, Huybrechts W, Ector J,Willems R, et al. Europace. 2008;10:351-7.2 - Fiabilidad técnica y seguridad clínica de un sistema de monitorización remota de dispositivos cardiacos antiarrítmicos.Joaquín Osca, María José Sancho Tello, Josep Navarro et al. RevEspCardiol. 2009;62:886-953 - What evidence do we have to replace in-hospital implantable cardioverter defibrillator follow-up?Brugada P. Clin Res Cardiol. 2006;95:3-9.4 - Remote monitoring and follow up of pacemaker and implantable cardioverter defibrillator. Haran B, Senouf D. Europace (2009) 11, 701–709.5 - Contributions of remote monitoring to the follow-up of implantable cardioverter–defibrillator leads under advisoryLaurence Guédon-Moreau, Philippe Chevalier, Christelle Marquie et al. European Heart Journal (2010) 31, 2246–2252 6 - Subclinical Atrial Fibrillation and the Risk of Stroke. Jeff S. Healey, Stuart J. Connolly, Michael R. Gold et al. N Engl J Med 2012: 366;2 7 - Asymptomatic atrial fibrillation: demographicfeatures and prognostic information from the Atrial Fibrillation Follow-up. Flaker GC, Belew K, Beckman K, etal. Investigation of Rhythm Management (AFFIRM) study. Am Heart J 2005;149:657-63.8 - Detection of atrial high-rate events by continuous Home Monitoring: clinical significance in the heart failure–cardiac resynchronization therapy population. Nesan Shanmugam, Annegret Boerdlein, Jochen Proff. Europace 2012 14, 230–2379 - 2012 ACCF/AHA/HRS Focused Update of the 2008 Guidelines for Device-Based Therapy of Cardiac Rhythm Abnormalities A Report of the American College of Cardiology. Cynthia M. Tracy, Andrew E. Epstein, Dawood Darbar. Foundation/American Heart Association Task Force on Practice Guidelines. Circ. 2012;126:1784-800. 10 - 2012 ACCF/AHA/HRS focused update incorporated into the ACCF/AHA/HRS 2008 guidelines fordevice-based therapy of cardiac rhythm abnormalities: a report of the American College of Cardiology Foundation/American Heart Association Task Force on Practice Guidelines and the Heart Rhythm Society. Epstein AE, DiMarco JP, Ellenbogen KA, Estes NA et al: Circ. 2013;127: 283-35211 - ESC Guidelines for the diagnosis and treatment of acute and chronic heart failure 2012.The Task Force for the Diagnosis and Treatment of Acute and Chronic Heart Failure 2012 of the European Society of Cardiology. John J.V. McMurray, Stamatis Adamopoulos, Stefan D. Anker et al. The Task Force for the Diagnosis and Treatment of Acute and Chronic Heart Failure 2012 of the European Society of Cardiology. Developed in collaboration with the Heart Failure Association (HFA). European Heart Journal (2012) 33, 1787–1847.12 - Automatic home monitoring of implantable cardioverter defibrillators. Nielsen JC, Kottkamp H, Zabel M, Aliot E, Kreutzer U, Bauer A et al. Europace 2008;10:729–35. 13 - Remote, wireless, ambulatory monitoring of implantable pacemakers, cardioverter defibrillators, and cardiac resynchronization therapy systems: analysis of a worldwide database. Lazarus A. Pacing ClinElectrophysiol 2007;30:2–12.14 - Cardioverter-Defibrillators Presenting to Nonelectrophysiologists. Clinical Assessment and Management of Patients With Implanted. William G. Stevenson, Bernard R. Chaitman, Kenneth A. et al. Circ. 2004;110:3866-386915 - Efficacy and Safety of Automatic Remote Monitoring for Implantable. Device Follow-Up (TRUST) Trial. Niraj Varma, Andrew E. Epstein, Anand Irimpen et al. Cardioverter-Defibrillator Follow-Up : The Lumos-T Safely Reduces Routine Office Circ. 2010;122:325-332;16 - Combined heart failure device diagnostics identify patients at higher risk of subsequent heart failure hospitalizations: results from PARTNERS HFPARTNERS Study Investigators. (Program to Access and Review Trending Information and Evaluate Correlation to Symptoms in Patients With Heart Failure) study. Whellan DJ, Ousdigian KT, Al-Khatib SM, Pu W, Sarkar S, Porter CB, Pavri BB, O'Connor CM; J Am CollCardiol. 2010;55:1803-10.17 - Failure Patients With Implantable Defibrillators : The Evolution of Management. Remote Monitoring Reduces Healthcare Use and Improves Quality of Care in Heart.|Strategies of Heart Failure Patients With Implantable Defibrillators (EVOLVO) Study. Maurizio Landolina, Giovanni B. Perego, Maurizio Lunati et al. Circ. 2012;125:2985-299218 - 2012 EHRA/HRS expert consensus statement on cardiac resynchronization therapy in heart failure: implant and follow-up recommendations and management. Daubert et al. Heart Rhythm 2012;9:1524 –1576.19 - HRS/EHRA Expert Consensus on the Monitoring of Cardiovascular Implantable Electronic Devices (CIEDs).Description of Techniques, Indications, Personnel, Frequency and Ethical Considerations. Wilkoff et al. Europace (2008) 10, 707–725.20 - Rationale and design of the health economics evaluation registry for remote follow-up: TARIFF. D'Onofrio A et al. Europace. 2012;14:1661-5.
Hugo del Castillo-Carnevali, Antonio Hernández-Madrid, Roberto Matía, and José Luis Zamorano.Cardiology Department. Ramón y Cajal Hospital. Madrid. Spain.Correspondence:Antonio Hernández-MadridArrhythmia UnitRamón y Cajal HospitalCarretera Colmenar Viejo, km 9,10028034 Madrid (Spain).Authors' disclosures: none described.You may find more articles on this topic by clicking on the following links: Remote monitoring for AF in EuropeSurvey on use of CIEDsDescription of devices and implantation Reimbursement and legal aspectsMonitoring in chronic heart failure Monitoring syncope and palpitations